Unfold the delivery arm, extend to horizontal and position as needed.
STEP TWO Loading Procedures
(Start up & replenishment only)
With the BS4000 on press the CODE button (center top row) ont he control panel. This will enable the CODE ENTERING MODE. Write down the units serial number that appears. Press CONTINUE and enter the 11 digit authorization code. The display will confirm receipt. Press EXIT.
You can now begin a BriteSmile chairside whitening procedure.
STEP THREE Reading the Keypad
The total number of procedures will appear on the display after the transaction is complete.
Options for starting a NEW procedure or a TOUCH UP appear just below the listing of total procedures available.
Once a new procedure is selected, the time remaining will appear on the display.
STEP FOUR Curing Light
Prepare the patient, following the clinical guidelines treatment protocol using the curing light inside the compartment.
STEP FIVE Start the Treatment
5.1 The display will show the current balance.
5.2 Select "New Procedure" to do a whitening procedure. (3 x 20 minute cycles).
5.3 Select "Continue" to proceed, "Abort" to cancel, and return to the main menu.
5.4 Once "Continue" is selected, the display will show the time remaining.
STEP SIX Position the Delivery Head
6.1 Grasp the delivery head with one hand and bring it close to the patient's mouth.
6.2 Use your other hand to raise or lower the arm to position the delivery head over the patient's teeth.
6.3 Use both hands to carefully "plug" the slender metal post on the delivery head into the hollow end of the optical positioner.
6.4 Have the patient bite down as the two parts join.
6.5 You will feel a slight "bump" as the delivery head plugs into the positioner.
STEP SEVEN Activating the Light Source
7.1 Press the "Start" button to activate the light source.
The "Pause" button can be pressed at any time to turn off the light.
Press the "Continue" button to restart the light and resume treatment.
7.2 The control panel will indicate the current session number and time remaining.
7.3 Press the "Start" button to activate the light between gel applications.
STEP EIGHT Completing the Session
(Fourth session option)
8.1 At the end of the third session, the display will read "Treatment Complete".
The operator can press "Done" to reset the device for the next patient or press "Extend" to go a fourth 20-minute session.
8.2 Press the "blue on/off" button to turn the unit off.